Difference between revisions of "PhoXi 3D scanners family"
(→Retrieving intrinsic parameters of the scanner)
|Line 139:||Line 139:|
== Retrieving intrinsic parameters of the scanner ==
Revision as of 15:36, 17 October 2018
This page contains technical description of PhoXi 3D Scanners. For scanning guide, see Scanning best practices.
|Light source||Visible red light (laser)|
|Laser class||3R (IEC / EN 60825-1, 2014)|
|Peak / CW Power||314uW|
|Projection angle horizontal||47.5° ±1°|
|Projection angle vertical||36.0° ±2°|
For model/size specific parameters see our product showcase.
Specification Parameters Explanation
- Depth Map resolution
- Maximum number of rendered points (the resolution of camera sensor)
- Point size
- The average distance between two neighboring points on the point cloud of a plane located in the focus distance of the camera. Alternatively, the square of the point size is the average surface sampled by one 3D point on the plane scanned in the focus distance of the camera.
- Calibration accuracy
- The accuracy of point measurement as the result of device calibration (can be understood as the space deformation). It is the standard deviation of the measurement error in the whole measurement range of the device.
- Calibration accuracy is measured by scanning multiple planes at different distances from the scanner. Below you can see a scan of the plane done by the PhoXi 3D Scanner model S at the distance of 411mm. The red points are closer to the plane and the blue points are further away form the plane
- Calibration accuracy is the average distance of the 3D points from this plane.
- Temporal noise
- The standard deviation of the noise (measured on a diffuse surface with 80 percent albedo). The noise level describes the quality of the sensor to capture local surface details. The noise distribution of our sensor is similar to the Gaussian distribution.
- Below is a graph depicting the Standard deviation of depth (Z-noise) values of 3D points for different scanning distances for the PhoXi 3D Scanner model S.
- Equivalently, Temporal noise can also be defined as the average distance of the 3D points from the average Z-value of the 3D points.
- Maximum number of triggered frames per second, in fastest acquisition mode
- Data acquisition time - best case
- Fastest possible acquisition time
- Data acquisition time - worst case
- Longest expected acquisition time (bad lighting conditions, dark shiny objects)
- 3D points throughput
- Number of 3D points that can be reconstructed in a second in sequential scans
|Status LEDs meaning|
| ⚫ Power status
Power cable connected
| ⚫ Processing unit connected|
On => Suitable power connected
| ⚫ Reserved
| ⚫ Reserved
| ⚫ Ethernet speed
Off => Indicates 1 Gbps
| ⚫ Ethernet activity|
Off => Link is down
Contains 3D Scanner body and its scanning volume as a solid object.
Available CAD models for download:
|Device||Mounting space coordinate space||Camera coordinate space|
|PhoXi 3D Scanner XS||XS - mounting space||XS - camera space|
|PhoXi 3D Scanner S||S - mounting space||S - camera space|
|PhoXi 3D Scanner M||M - mounting space||M - camera space|
|PhoXi 3D Scanner L||L - mounting space||L - camera space|
|PhoXi 3D Scanner XL||XL - mounting space||XL - camera space|
|All in mounting space||All in camera space|
Note: Camera coordinate space is only approximate.